- Electrical Instrument
- Sonic Toothbrush
- Digital engine analyzer
- Environmental Meters
- Non-contact voltage tester
- Digital tachometer
- Insulation Resistance Testers
- Ground resistance meters
- Other electrical test equipment
The use of the anemometer
anemometer probe select 0 to 100 m/s of velocity measurement range can be divided into three sections: low speed: 0 to 5 m/s; Medium: 5 to 40 m/s; High speed: 40 to 100 m/s. Thermal probe anemometer is used to 0 to 5 m/s measurement; Anemometer turn wheel probe measurement 5 to 40 m/s the velocity effect of zui ideal; While using pitot tube can be within the scope of the high-speed zui better results are obtained. Correct selection of the flow rate of the anemometer is an additional standard temperature of the probe, the use of thermal sensors usually anemometer temperature of about + - 70C。 Special anemometer wheel probe of up to 350 c. Pitot tube is used for more than + 350 c.
anemometer thermal probe
the working principle of thermal probe anemometer is based on the impact of cold air take away heat, heat element with the aid of a switch, to maintain a constant temperature, the adjusting current and flow rate is proportional to the relationship. When used in turbulent thermal probe, from all directions of air flow and heat shock element, which would affect the accuracy of the measurement results. Measurement in turbulent flow, the thermal anemometer velocity sensor is often higher than the value of the wheel probe. The above phenomenon can be observed in the process of pipe measurement. According to different design management of turbulent flow in pipe, even at low speeds. Therefore, anemometer measurement process should be performed in the linear section of the pipeline. Starting point of the straight line part should be at least 10 before the measurement point; D( D = pipe diameter, the unit is CM) Outside; The finish at least after the measurement point 4; D. Fluid section shall not have any shade. ( Edges and corners, the heavy suspension, content, etc. )
the propeller anemometer probe
the propeller anemometer sensor working principle is based on the rotation is converted into electrical signals, after a close first induction, beginning to count the rotation of the wheel and generate a pulse series, then through detector conversion processing, can get the speed value. Large diameter probe anemometer ( 60mm,100mm) Suitable for the measurement, small velocity of turbulent flow, As in the pipeline export) 。 Anemometer small-bore probe is more suitable for measuring pipe cross section is greater than the adventure changed head transverse 100 times more air flow.
anemometer in the location in the air flow
the propeller anemometer probe the correct position, the adjustment is the air flow direction parallel to the axle. Gently turn the probe in air, the values will change. When reading zui large value, namely that the probe in the correct position. Measurement in the pipeline, pipe straight part of the starting point to the distance measurement point thrown so 0 xd, turbulent thermal probe of anemometer and the influence of the pitot tube is relatively small.
anemometer in the smoke gas exhaust measuring
vent will greatly change the distribution of pipe flow is relatively balanced state: high speed in free surface of the vent area, the rest part is low speed area, and generate vortex on the grid. According to the different design methods of grid, a certain distance in front of the grid ( About 20 cm) , stable air flow cross section. In this case, usually with large diameter wheel of anemometer measurement. Because of the larger diameter can carry on the average, to the uneven flow velocity and calculating the average on a wide range.
anemometer in porosity volume flow funnel is adopted to improve the measurement with pitot tube flowmeter - - - - - - - - - - - - - JX1000 - 1 f intelligent pressure wind speed air flow meter:
even if without the distraction of grid on the extraction, air flow route are also have no direction, and the uneven airflow cross section. Partial vacuum, the reason is that the pipe to funnel put out in the air chamber in the air, even in the areas close to the suction also does not have the position of a measuring conditions, for measuring operation. If use with the function of average grid measuring method to measure, and measure so as to determine the volume flow method, and so as to determine the volume flow, etc. , only a pipe or funnel method can provide a repeatable measurements. In this case, different size of measuring funnel can satisfy the use requirement. Measuring funnel can be used to a certain distance in front of the plate valve in generating a fixed section, velocity measurement condition to measure the section center and fixed section, to measure the section center and fixed cross section, the location of the cross section is measured and fixed in the center. Velocity probe measurement value multiplied by funnel coefficient, volume flow rate can be calculated out. ( Such as coefficient of funnel 20) 。